JBW-H Series Instrumented Charpy Impact Testing Machine
|FOB Price:||Get Latest Price|
|Min. Order:||1 Piece|
|Production Capacity:||500 Sets/Year|
|Payment Terms:||L/C, T/T|
- Model NO.: JBW-H
- Maxcapacity: 750j
- Load Way: Electronic Load
- Display: Digital
- Weight: 100-500Kg
- Trademark: UnitedTest
- Specification: ISO, ASTM
- HS Code: 9031809090
- Type: Impact Testing Machine
- Accuracy Grade: 1
- Loading Method: Static Load
- Control: Computer Control
- Power Source: AC380V
- Transport Package: Standard Export Wooden Case
- Origin: China
This tester is mainly consist of automatically high and low temperature device (optional), auto-feeding system, and impact tester loadframe, can be used test, analysis the accuracy material anti-impact performance. Tester is controlled by computer, rising pendulum, hang pendulum, impact, stop all controlled by PC.
Adopt data collection system, change the voltage signal to digital signal, add data storage and PC, to transfer collected data to impact load - time, impact load-displacement, displacement- time, energy- time etc. Curve. Vivid illustrate the material under loading and deformation process to break.
1, Comply with GB/T19748-2005, ISO 14556: 2000(E), ASTM E23 etc.;
2, Loadframe base and leading column integrated design, overall process, with high stiffness, and improve test stability;
3, Impact axis adopt Charpy type, reasonable the radial direction load, maximum decrease energy loss caused by the bearing friction;
4, Adopt two level standard deceleration motor to rise the pendulum, rising and hang pendulum adopt hydraulic buffer to smooth the action;
5, Pendulum is round shape swing type, ensure the strike center accuracy, and precision pendulum moment.
6, Can achieve parameter of Impact velocity, impact time, impact energy, absorb energy, displacement, crack forming energy, crack extension energy, yield load, yield time, peak load, time of peak load, displacement of peak load, energy of peak load, unstable crack extension start load, unstable crack extension stop load, unstable crack extension start energy etc.
7, Working principle: Through the angle displacement sensor to measure pendulum pre-rising angle and rising angle after break specimen, then obtain tester impact energy and remain energy after specimen break, and supply each sampling point velocity parameter; Pendulum impact break the specimen at one time, the load sensor will measure the load changement during this period, and figure the load-time curve. Through software analysis load-displacement and other data, to help customer study the specimen deformation and breakage character.
8, Safety pin; And all close safety cover.
|Distance between pendulum center to impact point||750mm|
|Pendulum preparing angle||150º|
|Span of specimen seat||40+0.2mm|
|Round angle radius of Grips||R1-1.5mm|
|Round angle radius of striking edge||R2-2.5mm, R8± 0.08mm|
|Impact blade thickness||16mm|
|Angle of impact knife||30°|
|Load sensor accuracy||≤ ± 1%FS|
|Angle displacement resolution||0.1°|
|Lower temperature mode||Liquid nitrogen|
|Higher temperature mode||Heating wire|
|Sample box capacity||6 pcs, uniform 4 pcs|
|High and low temperature||Available and Optional|
|Sample Feeding system||can store 20 pcs sample at one time|
|Power supply||3 phases 4 lines, 50Hz, 380V|